For reagent free systems, users only need to add deionized water, and the system electrolytically generates the eluents and regenerants required for IC applications. The eluent generation technology results in a large closed loop system, reducing the need for external introduction of solutions into the system, which in turn reduces carbonate contamination. This can be one of the largest sources of instability and noise in IC. Further, the use of polymer and plastic pumps reduces the presence of ions from corrosion of metal components within the system.
To detect ions as they emerge from the column, particularly ionic species that are present at low levels, a conductivity detector with a wide dynamic range and high sensitivity to small changes in signal in a high ionic strength background must be used. However, it is typically simpler to convert the salts back to a slightly ionized form, so that any given ion will produce a more significant change in conductivity. This can be accomplished through a suppressor process, in which the eluent is passed through a second ion-exchange column.
What types of chromatography do you use in your lab? Start the discussion below!